The Intoxilyzer , while utilizing case law documented infrared spectroscopy as its method of analysis, is the first breath alcohol instrument in the industry to. The term Breath Testing Device (BTD), as used in this manual, refers to both the. Intoxilyzer (I) and the Intoxilyzer EN (EN). The terms are. (B) Prepare manuals and conduct courses throughout the state for the training of . In order to receive a permit to operate the Intoxilyzer breath alcohol.
|Published (Last):||5 March 2009|
|PDF File Size:||16.81 Mb|
|ePub File Size:||3.22 Mb|
|Price:||Free* [*Free Regsitration Required]|
The Department lnspector will submit all amended agency inspection reports or written explanations to the Program Manager.
This data is reviewed for the following:. The alcohol reference solutions must be from a different manufacturer than the solutions used in Department lnspections.
Table Of Contents – Lion INTOXILYZER Operator’s Manual [Page 3]
The Department lnspector is responsible for the receipt of amended documentation or requested written explanations as indicated on the Agency lnspection and Other Electronic Data Review document. The instrument will conduct four analyses of the standard. The five alcohol concentrations are:. Safety All required safety measures in place for the protection of personnel and equipment shall be observed.
The Program Manager or designee will review breath test electronic data at least once each month.
Lion INTOXILYZER 8000 Operator’s Manual: Table Of Contents
Department lnspectors will review cylinder change, login records, control test and diagnostic test electronic data of their assigned evidentiary intoxlyzer at least once each month. After all the solution values have been sampled, the auto cal data will be printed.
The Department lnspector will submit any amended documentation or written explanations to the Program Manager.
The Department lnspector shall review the inspection data in its entirety and confirm the following:.
The instrument will conduct four analyses of each standard attached. The Department lnspector should complete an Agency lnspection and Other Electronic Data Review document for any discrepancy determined from their review, send a copy to the agency inspector, and submit this document to the Program Manager.
Department lnspectors will be responsible for ensuring any requested documentation, when applicable, is received based on the review. Any action in reference to the instrument will be prepared in writing and sent to the agency inspector.
Any action in reference to operator training will be prepared in writing and sent to the breath test operator via the agency inspector. The results of the samples will be printed after the last sample is collected for that particular solution value.